The SpiroCard turns an off-the-shelf computer into a full-function spirometer. It exceeds the ATS Standard and out-performs its competition. The SpiroCard is pressure-calibrated at the time of production, so there is no need for routine calibration, and interfaces with Office Medic, bringing diagnostic testing right to the point-of-care.
The SpiroCard meets the following Standards:
- ATS/ERS 2005 Standard
- British Thoracic Society (BTS) Standard
- NICE Guidelines
- The SpiroCard was validated by the LDS Hospital’s Pulmonary Laboratory against 26 demanding waveforms and out-performed systems costing 10 times its price.It was found to have unparalleled accuracy and exceeds the ATS Standard.
Parameter ATS QRS SpiroCard
FVC Accuracy 3.0% Less than 0.2%
FVC Precision 3.5% Less than 1.3%
FEV1 Accuracy 3.0% Less than 0.04%
FEV1 Precision 3.5% Less than 0.9%
PEF Accuracy 10.0% Less than 2.0%
PEF Precision 5.0% Less than 0.8%
According to LDS Hospital Independent Test 2000; Based upon ATS Standardization of Spirometry 1994.
Included with each SpiroCard purchase:
- Pre-Calibrated Mouthpieces, 2
- Pressure Tubes, 2
- USB Adapter
- Office Medic Software